In this post you will learn about an approach to fabricate different stylet shapes for our proposed stylet and tube steerable needles using heat treatment of the Nitinol (a Shape Memory Alloy SMA).
Our lab developed a new class of steerable needles called Stylet and Tube Steerable Needles in which the direction of the tissue fracture is controlled by the inner Stylet followed by the outer tube to help steer the needle.
The shape of the stylet and the step length determine the radius of curvature of the steerable needle. For this reason, we have developed a method based on heat treatment of the Nitinol to make different shapes for the stylet and thus achieve different radii of curvature.
The contributors to this project are Fan Yang, and Mahdieh Babaiasl under the supervision of Dr. John Swensen.
Procedure for Heat Treatment of Nitinol Wires to Obtain Different Stylet Shapes:
To fabricate the optimal shape for each Nitinol stylet geometry, straight Nitinol pieces are pressed in an aluminum mold, heated to 500 C for 30 min, then quenched with water. Five stylet geometries are obtained after heat treatment.
The figures below show the step by step process to obtain different stylet shapes:
The same approach can be used to produce Nitinol wires with different pre-curvatures:
You can learn more about the Stylet and Tube Steerable Needles in the links below
If you enjoyed this post, please consider contributing to help us with our mission to make robotics and mechatronics available for everyone. We deeply thank you for your generous contribution!
Do not forget to contact us:
Be sure to let us know your thoughts and questions about this post, as well as the other posts on the website. You can either contact us through the “Contact” tab on the website or email us at support[at]mecharithm.com.
Send us your work/ research on Robotics and Mechatronics to have a chance to get featured in Mecharithm’s Robotics News/ Learning.
Follow Mecharithm in the following social media too: